Forgot your password?
No account? Register one!
March 15, 2010
Product SEN-08257 |
about 4 years ago
The sensor soldered on the breakout board needs to be reconditioned for accurate readings. From the SHT15 datasheet:
IMPORTANT: After soldering the devices should be stored at >75%RH for at least 12h to allow the polymer to re-hydrate. Otherwise the sensor may read an offset that slowly disappears if exposed to ambient conditions.
I did this by leaving my SHT15 in a ziploc bag with a cigar humidifier rated at rh 84%. You could also us a baggy with a small cup of damp salt. Before reconditioning the SHT15’s readings were -7% what they should be. Now it’s perfect. This is something I wish Sparkfun would do in house.
No public wish lists :(