Frequently Asked Questions
Mon-Fri, 9am to 5pm
U.S. Mountain Time:
Chat With Us
March 17, 2011
about 3 years ago
Does this allow debugging the baseline devices with just the pickit 2 (no ICD headers, etc)?
about 4 years ago
I agree about the OPTx pins. This is how the MN5010HS data sheet reads:
“The MN5010HS OPT1 and OPT2 input pins change the default behavior of the receiver upon startup
(either power-on or reset). Note that these signals do not have internal pull-up or pull-down resistors.
It is important to either pull up the input to 1.8 volts or pull it down to ground and not leave the lines
So why they are floating is a mystery to me.
There is a passive antenna already on the unit.
No public wish lists :(
Forgot your password?
No account? Register one!