Due to the impacts of the coronavirus outbreak, we are experiencing longer than normal lead times on certain products. We encourage back-ordering out-of-stock items to receive them as soon as possible.
Track My Order
Frequently Asked Questions
International Shipping Info
Mon-Fri, 9am to 12pm and
1pm to 5pm U.S. Mountain Time:
Chat With Us
November 7, 2011
Tutorial - Laser Tape Measure Hacking
about 8 years ago
Well, after reading through what you've provided (in terms of diagnostics and troubleshooting/tracing). I would have to guess that you've actually found the serial Rx pin on the chip (particularly if it traces back to one of the pads near the Tx test point pad. The likely problem, however, is that when the chip was built, it was probably done so with a fusing mechanism, so that after programming/testing/calibration is completed, the fuse is then 'blown' to lock that programming (there would be no need to blow the Tx pin, just the Rx). And voila you have a locked chip that prevents anyone later from communicating with the chip.
This may even be the same chip used in their higher end devices (which have serial connectivity available), and the 'fusing' mechanism would just be for use in the lower end units.
Please understand that this is all supposition and conjecture, though certainly within the realm of probability.
No public wish lists :(