SparkFun will be closed Monday 7/4 for the holiday. All orders placed after 2pm MT on Friday 7/1 will be shipped out next week.
Track My Order
Frequently Asked Questions
International Shipping Info
Mon-Fri, 9am to 12pm and
1pm to 5pm U.S. Mountain Time:
Chat With Us
March 1, 2012
about 4 years ago
From my observation, these pulse were NOT always scattered. Sometimes they were scattered but at other times a few pulse came to my board for a minute.
If you haven’t seen this problem with your test board, I want to know why I did and you haven’t. Environmental factor? Cs137 from Fukushima? (Definitely no. I confirmed that the place where my board sits is not contaminated by radioactive. - it is only 0.05uSv/h)
Anyway, with my modified firmware, any false pulses have been disappeared. (I should say “ignored” instead of “disappeared”.)
When I had this problem, the firmware of USB Weather Board running on my board was your v1.2 code.
I don’t change from RJ11 cables originally connected and don’t extend them,
so their length are probablly 10 fts long.
I’m sure that sitting of this rain gauge is very steady, because it is on the ground to examine what is the problem.
I modified the firmware to solve this problem.
I bought one and connected to my USB Weather Board. All sensors but Rain Gauge seem to work fine. I have a trouble with my gauge that it sometimes counts up without any rain. Yesterday, although there was no rain and humidity was less than 60%, it counted 25 times in a day!
Does anyone have the same problem?
No public wish lists :(
Forgot your password?
No account? Register one!