Track My Order
Frequently Asked Questions
International Shipping Info
Mon-Fri, 9am to 12pm and
1pm to 5pm U.S. Mountain Time:
Chat With Us
This product has shipping restrictions, so it might have limited shipping options or cannot be shipped to the following countries:
images are CC BY-NC-SA 3.0
added to your
Description: 36" high temperature Type-K Thermocouple with glass braid insulation.
Added to your cart!
This is the SparkFun USB to Serial Breakout for the FT232RL, a small board with a built in USB to serial UART
Detecting temperature changes has never been easier. The MCP9700 is a small thermistor type temperature sensor. This sensor w…
The SparkFun MAX31855K Thermocouple Breakout is a simple 14-bit resolution, SPI-compatible, serial interface thermocouple dig…
Melexis' MLX90614ESF-BAA is an infrared thermometer designed for non-contact temperature sensing. An internal 17-bit ADC and …
This Thermocouple Connector is perfect for bridging the gap between Type-K thermocouples with standard connectors with a PCB.…
This is a stainless steel, Type-K Thermocouple probe. A thermocouple works by taking two wires made of dissimilar metals, con…
Analog Devices Thermocouple Type-K Amplifier. Attach a thermocouple, 5V power, and a decoupling capacitor and you've got 10mV…
A row of headers - break to fit. 40 pins that can be cut to any size. Used with custom PCBs or general custom headers.
This is the new Arduino Uno R3. In addition to all the features of the previous board, the Uno now uses an ATmega16U2 instead…
It's blue! It's thin! It's the Arduino Pro Mini! SparkFun's minimal design approach to Arduino. This is a 3.3V Arduino runnin…
If you've ever tried to connect a 3.3V device to a 5V system, you know what a challenge it can be. The SparkFun bi-directiona…
This is the newest revision of our [FTDI Basic](http://www.sparkfun.com/commerce/product_info.php?products_id=8772). We now u…
Arduino is an open-source physical computing platform based on a simple i/o board and a development environment that implemen…
This sealed digital temperature probe lets you precisely measure temperatures in wet environments with a simple 1-Wire interf…
At SparkFun we use many Arduinos and we're always looking for the simplest, most stable one. Each board is a bit different an…
It's blue! It's thin! It's the Arduino Pro Mini! SparkFun's minimal design approach to Arduino. This is a 5V Arduino running …
This non-invasive current sensor (also known as a "split core current transformer") can be clamped around the supply line of …
This is a very small light sensor. A photocell changes (also called a [photodetector](http://en.wikipedia.org/wiki/Photodetec…
Single row of 40-holes, female header. Can be cut to size with a pair of wire-cutters. Standard .1" spacing. We use them exte…
The SparkFun Soil Moisture Sensor is a simple breakout for measuring the moisture in soil and similar materials. The soil moi…
Everyone knows and loves Raspberry Pi, but what if you didn't need additional peripherals to make it wireless. The Raspberry …
The SparkFun BME280 Atmospheric Sensor Breakout is the easy way to measure barometric pressure, humidity, and temperature rea…
An assortment of colored wires: you know it's a beautiful thing. Six different colors of solid core wire in a cardboard dispe…
A solid state relay (SSR) allows you to control high-curent AC loads from lower voltage DC control circuitry. Solid state rel…
Is this waterproof?
[sarcasm] Great for those projects where I, um, need to sense those greater-than-2500F… [/sarcasm]
Actually, no, because the glass braid insulation would melt before then (900 ºF).
So my guess here is that there is a tip at the end? I need maybe like 8" of the length for the project I’m working on, and I’d hate to waste the other 2+ feet of thermocouple wire, but from looking at the picture and reading comments I’m guessing that it might make my project more complicated if I try to shorten it myself. Am I right?
For people that will be reading this in the future… Do not shorten the wire! It may drastically decrease accuracy.
Please look up how thermocouples work before using one. I also recommend using a T-type instead, unless you need to measure extremely hot stuff.
AeroGuy is right. When you buy a thermocouple wire, the “sensing” part is just the very tip where the two metal wires are welded together. So if you cut the tip off, you’ll be left with just two metal wires. They can’t just be soldered together either. They must be welded together.
However, you can shorten the thermocouple at the other end, and it will work fine.
I have a problem with the K-type probe, the readings differ on LilyPad and UNO.
UNO has the temperature right, LilyPad 20 degrees less.
Any idea why is this happening? (of course the probe, code are the same)
Assuming you’re using their built-in ADCs, do they have the same reference voltage? I don’t remember if the ATmega328P has configurable internal reference voltage. Even if not, maybe one of them has a bad voltage reference circuit. Maybe also try using external reference voltage via the Aref pin—though I don’t know if the LilyPad breaks that out.
what is the max temp of the glass braid insulation?
Somebody mentioned a data sheet, someone else said “see below”, but I can’t find it…
potentially a stupid question, but would this work if it was encased in silicone?
Maybe this has been mentioned already but I left the tip of the sensor outside while it was raining, overnight. Now it’s not the same anymore. I can’t get accurate temperature readings as a consequence. So, I would not recommend using this in any situation with moisture.
can I know if this sensor can be used to measure the human body temperature plz?
You sure can, but the data will not be reliable since the reading may vary by (at least) 2-3 C, which is a substantial amount when it comes to human body temperature.
You can submerge type K- it is Nickle-Chromel (positive) and Nickle-Alumel (negative) Type T, J you might not want wet- especially J (Iron/ Constatan) the max upper temp limit for 30ga type K is 1600F- 18 and heavier is 2000
for protection use a ceramic tube- other metals may affect the alloy in type K as different trace elements diffuse off at higher temperatures
wikipedia has great info on type K (and thermocouple), but the book to look for super specifics on thermocouple is ASTM e-230
I work for a thermocouple facility (and actually supply omega!) any questions feel free to ask!
Is there a datasheet for this Thermocouple?
yeah sorry i made a general post- look below!
Anyone know what the accuracy specs are for this? My hope is to use this to calibrate our home ovens, which seem to always take longer to cook food than recipes indicate. (I’m at a low altitude. The ovens' documentation has a snobby comment about how they’re perfectly calibrated, thank you very much.)
yeah just use the thermocouple- the mfg has specific requirements it has to meet (EMF- eletro-motive force) (k cant vary more than .089mv from 200-500F)there are 32 mv/degree
if you are super concerned- look for a special limits sensor- it will cost a bit more but it is only allowed .044mv from 200-500
I need a sensor for use in a glass kiln that can achieve temperatures of up to 1800 deg. F. This sensor covers that range according to the specs, but the glass insulation is only good up to 900 deg. F. I don’t get it. Why would you take a sensor that is rated to 2501 deg. F. and cover it with insulation only good to 900 deg. F. Am I missing something? Does one remove the insulation for higher temp applications, or is this sensor really not usable above 900 deg?
at 2501F you are going to need a very very expensive thermocouple- a type R/B to get those very high temps-
the thermocouple needs to be the only thing exposed-after you use it you can slide it in a little further and it will maintain its accuracy
This is a k rated sensor, however the probe tip is a tiny affair, it’s the bit on the end of the insulation. On kiln temp sensors the tip is several inches long, this allows it to be inserted through the kiln wall without exposing the insulation to excessive heat.
You can find ones designed for kilns by searching “k temperature probe” on google
The short answer is probaly cost. The sensor functions at 2500 degrees but the materials may nit last long at high temperatures. Tcs made for that type of enviromemt are sheathed in high performance alloys. Most people won’t need that. For your application checkout omega.com, there website is not very good so you may want to call.
Not sure if this is going to be much help.
Many years ago I worked at a metallurgical test lab. They basically tested samples of metal under high tensional stress along with high temperature. There was a guy whose sole job was building thermocouples that would sit inside a furnace to measure the temperature of the specimen under test.
He would take two pieces of dissimilar wire and spot weld them at one end to form the hot junction. To insulate the two pieces of wire from each other and from the outside world, he would thread ceramic beads onto the two wires.
Each ceramic bead was cylinder shaped, about 6mm in height and about 4mm in diameter. There were two small holes running through the cylinder.
The finished thermocouple was very fragile and needed great care during installation. It resembled a necklace of pearls. The goal though was that it could stand extremely high temperatures and was somewhat flexible for routing out of the furnace.
I’m afraid that I have no idea where to source the ceramic beads, or the bare al/ch wire to make a K thermocouple from scratch.
Ditto to this question!
This looks like the same as this thermocouple from McMaster-Carr
Note: You cannot solder to the ends of this thermocouple.
I have this TC… has anyone else used this? I’m using the MAX6675 TC amp and the arduino. All seems to work well. However I would like to use this TC under water. I was thinking of epoxying the very end of this TC into a small diameter stainless steel tube, but I’m not sure how this will effect it’s accuracy. The temp range of the application is on the low side <200 degrees F. Any ideas? Would plastic work?
Stainless is a terrible conductor of heat. Epoxying this inside a SS tube would effective be insulating it from whatever you’re trying to measure.
Although stainless conducts heats orders of magnitudes slower than pure copper or aluminum, that would really only have an effect on the response time of the system. It will reach the steady state temperature of the solution, but it may not represent the temperature of the solution exactly if the solution’s temperature is varying rapidly. Stainless thermocouples are used all the time in both chemical and biological processes. Stainless still conducts much better than air. For comparison, aluminum’s thermal conductivity = 250, stainless steel’s thermal conductivity = 16, and air’s thermal conductivity = .024 W/(mK).
Any idea when it will be restocked? Can you say who its from?
Based on 1 ratings:
about 2 years ago
by Member #332504
The previous one had frayed, and it was a snap to put this new one on.
Forgot your password?
No account? Register one!